- LEAFLET & VIDEO
- Contact Us
MFI series Melt Flow Index Tester/Extrusion Plastometer.
They are suitable to measure the melt flow character of PP, PE, PVC, POM, ABS etc., plastic materials resin and granules.
Principle: The melt mass-flow rate (MFR) and the melt volume-flow rate (MVR) are determined by extruding molten material from the cylinder of a plastometer through a die of specified length and diameter under preset conditions of temperature and load.
Feature of MFI405:
PC and software control, both MFR/MVR test method, motorized cutting, show the temperature VS time curve, connect A4 paper printer to print test report, test data.
Approximately 7 grams of the material is loaded into the barrel of the melt flow apparatus, which has been heated to a temperature specified for the material. A weight specified for the material is applied to a plunger and the molten material is forced through the die. A timed extrudate is collected and weighed. Melt flow rate values are calculated in g/10 min.
At least 14 grams of material;
Flow rate = ( 600/t x weight of extrudate )
t = time of extrudate in seconds
melt flow rate = g/10 min.
Melt flow index testing equipment; melting flow indexer; melt indexer iso 1133; digital melt flow rate tester; quality melt flow index testing machine; melt flow index astm; plastic Melting Flow Index tester
|Temperature range||Ambient temperature to 400℃ stepless adjustable, fixed point temperature control|
|Weight||8 levels, total 21.6kg mass|
|Measuring method & range||MFR (0.1-100)g/10min|
|Extruding part spec. :|
|(1), Inner diameter of die||φ2.095±0.005mm|
|(2), Length of die||8.000±0.025mm|
|(3), Inner diameter of filling canister||Φ9.550±0.025mm|
Standard weights: 25kg
|Power||220V, 50HZ/110V, 60HZ|
|Display||Computer display temperature, time, sampling number, temperature VS time curve etc.|
|Specimen cutting mode||Automatic & Manual both mode|
|Other model: |
1, MFI401 Digital display, economical type, manual cutting, only MFR test method;
3, MFI403 LCD display, motorized cutting, both MFR/MVR test method;
ModelMFI401MFI402MFI403MFI404MFI405Test methodMFRMFRMFR/MVRMFR/MVRMFR/MVRDisplayDigitalLCDLCDTouching screenComputerPrinterNoMicro-printerMicro-printerMicro-printerA4 paper PrinterSpecimen cuttingManualAutomaticManual/AutomaticManual/AutomaticManual/AutomaticTest curveNoNoNoNoYes?
|Die (φ2.095) tungsten carbide material||1 pc|
|No. 1 weight (piston rod + weight tray): 325g||1 pc|
|No. 2 weight: 875g||1 pc|
|No. 3 weight: 1835g||1 pc|
|No. 4 weight: 3475g||1 pc|
|No. 5 weight: 4675g||1 pc|
|No. 6 weight: 5000g||1 pc|
|No. 7 weight: 2500g||1 pc|
|No. 8 weight: 2915g||1 pc|
|Sample loading hopper||1 pc|
|Water level||1 pc|
|Die cleaning rod||1 pc|
|Cylinder cleaning rod||1 pc|
|Sample loading rod||1 pc|
|Sample plate||1 pc|
|Electric wire||1 pc|
|Cleaning gauze and gloves||1 pc|
|Accessories containing case||1 pc|
|Documents (Manual, packing list, certificate)|
ISO 1133 "Plastics - Determination of the Melt Mass-Flow Rate (MFR) and Melt Volume-Flow Rate (MVR) of Thermoplastics”
ASTM D1238 "Standard Test Method for Melt Flow Rates of Thermoplastics by Extrusion Plastometer“
ASTM D3364 (for PVC), BS 2782, DIN 53735, JIS K7210